is fixed properly, and that the legs do not move inside working position. If you are not able to solve the fault or problems by yourself using this manual, please contact local authorised service representative. Nominal battery voltage Molift Smart 150 short guide for mounting Model name read user manual! Check that the emergency stop button is Lifting arm Fig A not activated and that the lifter is functioning. Press the emergency lowering button. The owner is responsible to ensure proper logging and written verification of each inspection. The lifter shall only be used for movement over short distances. If taken into consideration, it will help the operator of the lifter to work efficiently. Ensure that arms and legs are clear of the lift, chair, bed etc. 3 General Safety precautions.4 General.
Take contact with your local service representative and order service. Fill in and sign a copy of the checklist on next page. Periodic inspection Etac Supply Gjvik recommends periodic inspection once a year, alternatively more often if required by local regulations.
Cleaning Detergents must be ph-neutral. Remove hair and pile from the castors and verify that the castors rotates without abnormal friction. No: Hard case for Molift Smart Part. The control electronics has failed / contact local representative for service and use manually emergency lowering, where necessary Short circuit between hand control and lifter / clean hand control plug surfaced with alcohol to remove grease Defective hand control / replace hand control / contact. Be particularly careful when manoeuvring close to furniture to prevent the suspended user from colliding with these objects. Remove the transport elastic and spread the legs to working position (maximum width). Comments on faults and repairs: Lifter serial.: Other. The Molift Smart 150 with accessories as described in this manual are CE marked according to the Council Directive concerning medical devices (93/42/EEC). Molift Easy Sling Size No head support With head support XS Art. BM09201 Molift smart 150 English - Rev E 10/2012 Page 14 of 20 15 Troubleshooting Faults on lifters that do not function as they should after assembly (page 5) or after the pre-use check has been completed (page 7 can be localized using the troubleshooting.